These are the starting conditions of the Calibrate Manually tab following simulated annealing of the radionuclide source strength distribution in the Surface Measurements tab.
Begin by adjusting the Dose Calibration slider until the dose profile curve passes through the 5 mm measurement point.
Note: To remove the bump in the curve between 0 and 0.25 mm, increase the nearfield slope and then adjust the crossover and nearfield slope sliders keeping the curve between 0 and 1 mm as smooth as possible by inspection.
RMS % error (0 to 10.0 mm) = 3.160
Repeat steps 2-8 until the RMS error over the range 0..10 mm can no longer be reduced.
RMS % error (0 to 10.0 mm) = 1.383